A Low Supply Gas Pressure Alarm, now standard on all units, will sound when the supply gas pressure falls below 35 psi, 2.4bar and stops around 42 psi. The normal supply gas operating pressure is 50psi or 3.5bar. It is not until 35psi or 2.4bar that the ventilators delivered volume will begin to drop off proportionally with the waning supply pressure.
The single valve turns a total of nine and a half times; the first two and a half turns utilizing the slight, minute changes for pediatric animal patients, and the remainder of the rotations for use with the adult and extra large bellows and housings. All other parts and accessories for the Model 2002PRO have remained the same.
All 2002PRO series purchases now include our Ventilator Accessory Kit.
Electronic control-- Eliminates the substantial driving gas required to operate pneumatically controlled ventilators.
Time cycled-- Ensures consistently spaced breaths. The possibility of stacking breaths or auto-PEEP is eliminated by keeping the I:E ratio constant at 1:2. This means that regardless of the respiratory rate setting two-thirds of the breath will remain for expiration. The respiratory rate is adjustable from 6 - 40 breaths per minute (BPM).
Volume constant-- Ensures that each delivered breath will have essentially the same volume, independent of changes in animal patient compliance and airway resistance. The deliverable tidal volume (TV) range is from approximately 20 cc to 3,000 cc.
Looks like you haven't made a choice yet.